The two graphs above show water level height and percent new water that entered each river during Hurricane Sandy on October 30, 2012. The Israel River's peak in water composition lagged behind the peak in water level by 4 hours and 12 minutes. Keene's Beaver Brook, lag time between peaks was only 42 minutes. One hypothesis for this difference is that urban surfaces are less permeable than most of the surfaces which they replace, so rain has less interaction with soils in urban surfaces. Check back for other comparisons. I will be running analysis and posting new graphs.
Monday, December 10, 2012
|Hobo U24 Datalogger configuration in Blanchard Brook pool|
|Left-Right: Hobo U24 Datalogger, YSI 85, Oakton EC 11+|
Early in November, we installed LoVoTECS Hobo Water Level and Conductivity Loggers in a small stream called Blanchard Brook in Walpole, NH. The loggers were installed as part of a quality control group to study the variations among the dataloggers. The installation method differs from the rest of the network, in that five conductivity loggers and one water level logger were deployed in a one foot section of a pool in Blanchard Brook. The loggers are collecting data points every minute rather than every three or fifteen. Each data set will be compared to identify variations among individual loggers. Since we’ve been observing the electrical conductance of Blanchard Brook, the average conductivity range has been between 100 and 200 us/cm.
In addition to the comparison of the individual units, we are also capturing field conductivity readings using two styles of field meters and collecting a grab sample at the time of the field readings to note the variation between field, lab and the datalogger values.
Following the deployment in Blanchard Brook the logger group will be moved to Claybrook in Plymouth to make the same observations. Claybrook has an average conductivity range from 30-60 us/cm. Finally the loggers will be placed in a series of three standards to test their accuracy against the known solutions in a controlled environment.
Stay tuned to the blog to follow the progress and results of the Quality Logger Group.